Easy Way To Calibrate a Bryman BM869s Multimeter
After replacing my lost Fluke 87V multimeter with a new Brymen BM869s (rebranded Greenlee in the US) about a year ago, I’ve been delighted with the multimeter. It has every possible function I could ask for and then some. It’s been my go-to meter when working on electronics projects. I do have a Siglent SDM3055 DMM that is very good, but due to its startup time, I usually reach for my Brymen BM869s when I need to measure something, and it has higher resolution. But after getting a broken HP/Agilent 34401A DMM from eBay, fixing it up and replacing all the capacitors, I now have a better reference. I’ve compared it to a friends calibrated 6 ½-digit multimeter, and it’s still within spec, so naturally, I want my Brymen to be as close to my HP 34401A as possible. Turns out that calibrating and adjusting the Brymen is quite easy. Here’s how to do it.